with ToggleLoc™ Fixation Device with ZipLoop™ Technology and ComposiTCP™ Interference Screw
ComposiTCP Interference Screw
The composite blend of 40% PLDLA and 60% beta Tri-Calcium Phosphate has been studied and the results reported in an in-vivo animal study1 showed that «in comparison with pure PLA, TCP-containing composite materials had faster degradation kinetics, caused less inflamatory reaction, and promoted contact osteogenesis.» In addition, an in-vitro study discussed how increased amounts of TCP have been shown to stimulate the proliferation of osteogenous cells1
Utilizing a tibial guide that allows for optimal tunnel placement, position the tibial guide appropriately and drill the guide wire. After the graft size has been determined, ream over the guide wire with the corresponding reamer. Position a Femoral Aimer into the over-the-top position through an accessory anteromedial portal (Figure 1)
This material represents the surgical technique utilized by Jefferey Michaelson, M.D. Zimmer Biomet does not practice medicine. The treating surgeon is responsible for determining the appropriate treatment, technique(s), and product(s) for each individual patient.
Tunnel Preparation (cont.)
Drill a calibrated guide wire through the Femoral Aimer and the lateral cortex of the femur (Figure 2). Consider placing the scope into the standard medial portal to check that the guide wire is placed in the 9:30 – 10:30 position for a left knee and a 1:30 – 2:30 position for the right knee.
Drill over the previously placed guide wire an endoscopic reamer corresponding to the diameter of the graft diameter and ream to the depth that will allow the desired soft-tissue graft-to-tunnel interface (typically around 25 mm) (Figure 3). Drill over the previously placed guide wire with the 4.5 mm ToggleLoc drill bit through the lateral cortex of the femur (Figure 4). Pass the 4.5 mm drill in and out of the cortex two to three times to facilitate passage of the implant.
Tunnel Preparation (cont.)
Pass the ToggleLoc depth gauge into the femoral tunnel and measure the tunnel length from the lateral cortex of the femur to the tunnel exit point in the joint space (Figure 5). Pass the soft tissue grafts through both loops of the ToggleLoc Femoral Fixation Device with ZipLoop Technology (Figure 6). The implant should be left in the white cardboard packaging. This will facilitate passing the soft tissue graft through the correct loops. Place the graft through the hole in the package. Balance the soft tissue grafts in the loops of the implant to allow equal amounts of the soft tissue on either side of the loop.
Use the measurement previously obtained with the ToggleLoc depth gauge to mark the loops of the implant to ensure deployment on the lateral cortex. Measure from the distal end of the ToggleLoc device toward the graft and mark the length (Figure 7). Make a second mark on the graft by measuring the depth of the “graft tunnel” (typically 25 mm). This mark will aid in optimal graft positioning later in the procedure.
Thread a strand of relay suture through the eyelet of the graft passing pin so that the suture forms a continuous loop (Figure 8). Pull proximally on the guide wire to pull the relay suture through the skin. Use a suture grasper or crochet hook to retrieve (Figure 9) the relay suture through the tibial tunnel (Figure 10).
Loop the passing suture (white #2 suture pre-loaded into the titanium button) of the ToggleLoc Femoral Fixation Device with ZipLoop Technology through the relay loop, which should be exiting the tibial tunnel. Pull proximally on the relay suture to pull the passing suture through the tibial tunnel, joint space and femoral tunnel, exiting through the skin.
Insert Implant into Tunnel
Prior to fixation, ensure that the ToggleLoc Femoral Fixation Device with ZipLoop Technology is oriented laterally, as it will deploy on the femur’s lateral cortex. The “zip suture” should be on the anterior side of the soft-tissue graft prior to graft placement within the femoral tunnel (Figure 11).
Pull the passing suture proximally until the mark on the loops of the ToggleLoc device reach the entrance of the femoral tunnel. Position the implant just beyond the the lateral cortex of the femur (Figure 12). Pull on the distal end of the soft tissue grafts to feel the implant engage on the lateral femoral cortex, achieving femoral fixation.
Position Graft in Femoral Tunnel
Ensure the “zip suture” is anterior to the graft. Place the knot of the zip strand into the ziploop puller (Figure 13) and pull distally to draw the graftthrough the tibial tunnel and into the femoral tunnel. This will shorten the loop of the ToggleLoc Femoral Fixation Device with ZipLoop Technology and accurately position the soft-tissue graft in the femoral tunnel.
Correct placement is indicated when the mark on the graft enters the femoral tunnel. Cut the knot off of the end of the “zip suture” and retrieve the cut suture limbs through the medial portal (Figure 14).
Pass a 1.1mm nitinol guidewire through the tibial tunnel. Tap the tibial cortex if necessary and insert the desired ComposiTCP Interference Screw to achieve tibial fixation (Figure 15). If required, tension the femoral fixation by pulling on both limbs of the zip strand.
Sever the Zip Suture
Pass the limbs of the zip strand through the key shaped hole in the Super MaxCutter™ instrument (Figure 16). Advance the Super MaxCutter device through the medial portal and cut the suture at the entrance of the femoral tunnel in the joint space (Figure 17)
|904755||ToggleLoc Femoral Fixation Device with ZipLoop Technology|
|905256||9 x 25 mm||ComposiTCP Interference Screw 60% B-TCP—Round Head|
|905257||9 x 30 mm||ComposiTCP Interference Screw 60% B-TCP—Round Head|
|905258||9 x 35 mm||ComposiTCP Interference Screw 60% B-TCP—Round Head|
|905261||10 x 30 mm||ComposiTCP Interference Screw 60% B-TCP—Fully Threaded|
|905262||10 x 35 mm||ComposiTCP Interference Screw 60% B-TCP—Fully Threaded|
|905263||11 x 30 mm||ComposiTCP Interference Screw 60% B-TCP—Fully Threaded|
|905264||11 x 35 mm||ComposiTCP Interference Screw 60% B-TCP—Fully Threaded|
|909848||ToggleLoc Femoral Fixation Device with ZipLoop Technology Implant Kit|
|110007421||ToggleLoc Inline ACL Disposable Kit
3.2 Drill Bit
ToggleLoc Inline Implant
4.5 mm Rigid
ToggleLoc Drill Bit
1.1 mm Calibrated
Nitinol Guide Wire
Calibrated 3/32” x 16”
Trocar Tip Pin
Cannulated Bone Plug
|110004607||7-8 mm||Modular Driver|
|905274||9-11 mm||Modular Driver|
|900342||Super MaxCutter Suture Cutter|
|904760||4.5 mm||Drill Bit (Disposable)|
|904765||4.5 mm||Drill Bit (Reuseable)|
|904766||ToggleLoc Depth Gauge|
|11004136||ToggleLoc ACL Disposable Kit
2.4 mm x 13” Drill Point K-Wire
2.4 mm x 16” Graft Passing Pin ToggleLoc 4.5 mm Drill Bit
2.4 mm x 10” Drill Point K-Wire
1.1 mm x 14”
Nitinol Guide Wire
3.2 mm Drill Bit
ACL Bone Plug
|900716||Driver Bone Dowel Handle|
|900733||Driver Ratchet Handle|
|110004597||7-8 mm||Modular Tap|
|905278||9-11 mm||Modular Tap|
|905045||7-8 mm||Modular Dilator|
|905046||9-10 mm||Modular Dilator|
|906849||1.1 mm x 14”||Nitinol Wire|
|906852||1.1 mm x 9”||Nitinol Wire|
ToggleLoc Fixation Device
INDICATIONS FOR USE
The ToggleLoc System devices, except the ToggleLoc XL device, are intended for soft tissue to bone fixation for the following indications:
Bankart lesion repair
SLAP lesion repairs
Capsular shift/capsulolabral reconstruction
Rotator cuff tear repair
Foot and Ankle
Medial/lateral repair and reconstruction
Mid- and forefoot repair
Hallux valgus reconstruction
Metatarsal ligament/tendon repair or reconstruction
Achilles tendon repair
Ankle Syndesmosis fixation (Syndesmosis disruptions) and as an adjunct in connection with trauma hardware for Weber B and C ankle fractures (only for ToggleLoc with Tophat/ZipTight Fixation Devices)
Ulnar or radial collateral ligament reconstruction
Lateral epicondylitis repair
Biceps tendon reattachment
ACL/PCL repair / reconstruction
ACL/PCL patellar bone-tendon-bone grafts
Double-Tunnel ACL reconstruction
Extracapsular repair: MCL, LCL, and posterior oblique ligament
Illiotibial band tenodesis
Patellar tendon repair
Joint capsule closure
Hand and Wrist
Collateral ligament repair
Scapholunate ligament reconstruction
Tendon transfers in phalanx
Volar plate reconstruction
The ToggleLoc XL device is used for fixation of tendons and ligaments in cases of unanticipated intraoperative complications such as cortical breaching during orthopedic reconstruction procedures, such as Anterior Cruciate (ACL) or Posterior Cruciate (PCL) Reconstruction.
- Patient conditions including blood supply limitations, and insufficient quantity or quality of bone or soft tissue.
- Patients with mental or neurologic conditions who are unwilling or incapable of following postoperative care instructions.
- Foreign body sensitivity. Where material sensitivity is suspected, testing is to be completed prior to implantation of the device.
ComposiTCP 60 Interference Screw
INDICATIONS FOR USE
The ComposiTCP 60 Interference Screw is exclusively used for the fixation by interference of the transplant made out of soft tissue, taken out for instance from the hamstring tendon, when reconstructing the cruciate anterior ligament. The screws are cannulated and are available in different sizes (see commercial documentation). They have a specific head, which allows for a more even distribution of the torsional stresses. To achieve the optimal result, the ComposiTCP 60 Interference Screws should be implanted using a dedicated screwdriver contained in the instrumentation set.
Insufficient or poor-quality bone stock (including tumors and sever osteoporosis) is likely to affect screw purchase. Acute infection. Allerfy to implant material. Conditions likely to limit the patient’s ability and/or willingness to restrict activities and/or to adhere to instructions during the healing and rehabilitation period.